We do not sell the EasyOne World anymore. This product was replaced by the EasyOne Air. You find more information about the EasyOne Air on the product page.
FAQs EasyOne World/Plus
You can find the product specification on the respective product pages. The specification contains a list of all supported normal values.
To get more detailed information about the supported parameters, age range, ethnicities and height range of each supported predicted set, please click here.
BTPS Correction is used to convert flow and volume measured at ambient conditions to the conditions within the lungs. Ambient conditions are called ATP (ambient temperature, pressure); the conditions within the lungs are called BTPS (body temperature, pressure, water vapor saturated).
Click here for more information.
This error is usually a communication error shown while printing. It means something wasn’t quite ready. Typically, it is best to clear the error by hitting 1 then 3 then 5 on the keypad and to try again. Most Self-Test errors can be cleared and investigated with the code 1-3-5.
The EasyOne only works with a limited set of printers because it uses specific, aging Print Command Languages. Please check to see if your printer is on the Printer Compatibility List. If your printer is listed but the spirometer still won’t work, verify that the printer is not in error mode and check all points of connection with the EasyOne.
There are a few points that can be checked and improved. The EasyOne uses two AA batteries and, when these get low, communication may falter. Fresh batteries are best. Also, the 9-pin contacts on the bottom of the EasyOne and Cradle may oxidize. Polish these with an alcohol swab to clean them. Finally, the USB ports may be damaged. Check the cradle and printer or computer for signs of wear and try a new USB cord. The EasyOne uses a common type A/B USB cord.
You will need to use our software. You can download EasyWare 2013, which was created for the EasyOne, and use a PDF writer like CutePDF to activate PDF options, or you can use EasyConnect and create a PDF through the Report menu on a test synchronized to that program.
Yes, the EasyOne can synchronize with our EasyConnect software, which has EMR compatibility through HL7 and a number of custom interfaces. Since the EasyOne is a portable unit, the workflow would be that an order with patient data goes from EMR to EasyConnect to EasyOne. Then the test is done on the EasyOne and, when it is finished, that data goes from EasyOne to EasyConnect to EMR. This puts real data, not just a PDF snapshot, into the EMR and your patients’ charts.
Specifications EasyOne World/Plus
14971 , EN 62366 , EN 62304 , EN ISO
26782 , EN ISO 23747 , IEC 60601-1, IEC
and 10 (32 and 64 bit))
EasyWare Pro (Windows 7, 8, 8.1 and 10
(32 and 64 bit))
Rel. Humidity 0 - 95 %
Atmosph. Pressure 500 - 1060 hPA
MEF25, MEF50, MEF75, MMEF, PEF, PEFT, t0
FIF75, FIVC, FVC, MEF25, MEF50, MEF75, MIF25, MIF50, MIF75, MMEF, PEF, PEFT, PIF, t0
(Harvard) 1993, Polgar 1971, Gutierrez (Canada) 2004, Eigen 2002
(Mexico, Pediatrics) 2003, Chile 2010, Chile (Pediatrics) 1997
Finnish 1982 (1998), Nystad 2003